1 FREE box of needles with orders over $599!
The only you kit you need to remove a fractured or broken screw and save the implant. Based on the proven counter-torque and reverse screw technique (Froum S., Compendium. 2011;32(7):2-24).
Kit includes: 6 Guides for various implant systems (see below), 2 Drills, 1 Tap, 1 Guide Handle, 1 Remover Wrench. The components in this kit are identical to those supplied in the Salvin Implant Rescue Kit, which is a combination of the Fractured Screw and Implant Removal Kit.
Use the power of counter-clockwise force to effectively remove fractured screws with the this kit.
Fractured Screw Removal Kit Components
How to Use the Fractured Screw Removal Tools
Fractured Screw Removal Kit Components
2 Drills included in the kit
Please use the guide below for compatibility of the guides in the fractured screw removal kit.
Compatibility: Guides that Come with Kit
Hex Guide (H)
Octa Guide (O)
External Mini (M)
External Regular (W)
Other Guides Available for Purchase Seperately
3i Internal RP
Nobel Replace Tri-Lobe RP
Nobel Replace Tri-Lobe WP
Nobel Active NP
Nobel Active RP
|Questions and Answers|
Please refer to our [compatibility guide](https://www.ddsgadget.com/instructions/2010-fractured-screw-comp/).
Please refer to our [Fractured Screw Instructions](https://www.ddsgadget.com/instructions/2009-fractured-screw/).
Thanks for your question. You are correct, the Octa Internal guide will go with the Straumann Tissue Level implant, but not the Straumann Bone Level implant. For the Straumann Bone Level, please check out our Narrow Implant Broken Screw Removal Tool Kit (simply use the search box to search for this product or go to our Removal Kit section.
No. The kit will not work with small diameter or mini implants. The kit will only work with conventional implants. Usually, this means that the implants are 3.5mm in diameter or greater. However, on some systems, such as HiOssen, the mini implants are 3.5mm. On these systems, the kit will only work with the regular/conventional implants, that are 4.0mm or greater.
For the 3i Osseotite implant, you would use the External Regular (W) guide that comes with the kit. For the Straumann Bone Level implant, you can check out our Narrow Screw Removal Kit (simply search for it in the search box)
The tap drill is only used to engage the free loading broken screw, to get it out. So there should be very little torque applied to it. The tap drill must be used with very low speed 15-30rpm and not too much vertical pressure at all. If you use too much torque, and apply too much pressure, the tap drill can break.
No, it will not work with every fractured screw.
Firstly, it is important to make sure your implant system is compatible with this kit. See [the compatibility list](https://forums.ddsgadget.com/t/fractured-implant-screw-kit-compatibility/71).
Secondly, while our Fractured Screw Removal Kit is very popular, and works extremely well, it will NOT work in every single instance of a fractured screw. The kit is primarily indicated when the broken screw is fractured in the apical 50% of the implant chamber. In other types of fractures, other instruments maybe needed to remove the fractured screw. To read about the various methods of removing a fractured screw, [we recommend this article](https://www.perioimplantadvisory.com/clinical-tips/article/16412202/fractured-dental-implant-screw-complications-3-methods-for-screw-retrieval%20) (our fractured screw removal kit is featured as option 3, under the Salvin Implant Rescue Kit. The Salvin Implant Rescue Kit is the same exact thing as our Fractured Screw Kit and Implant Removal Kit. It is simply a combination of both of those kits in once kit).
Yes, you can order replacement drills. The Screw Removal Drill is SRD-01 and the Tap Drill is SRT-01. Just search for SRD-01 or SRT-01 or replacement drill in the search box and you can order those replacement drills.
Nobel Biocare’s implant systems are quite confusing. But, essentially, there are 3 different connections available: Tri-Lobe, Conical Connection (CC), and External. The External connection is used on the original Branemark implant line. The CC and Tri-Lobe are used on NobelReplace, and CC is used on the Nobel Active. Our kit includes guides that will work with the External connection for the Branemark (both Narrow platform and regular platform). The Kit also includes a guide that will work with the Trilobe Connection for NP platforms (3.5). However, for TriLobe connections for RP, and WP, you will need to purchase additional guides that are available thru our website. As for the Conical Connection (NobelActive and NobelReplace), these are not included in the kit, and need to be purchased separately thru our website. We only supply CC guides for the NP (3.5) , and RP (4.3, or 5.0).
Video for Fractured Screw Removal Kit
This video provides a short demonstration of the fractured screw removal kit and shows the kits components.
DDSGadget Guide for:
Fractured Screw Removal Tool Kit